This construction of battery powered dual face radiographic viewing box with rotatable neck and adjustable stand was carried out within Enugu metropolis.
TABLE OF CONTENTS
Table of contents-------iv
Background of study-------1
Statement of problem-------3
Purpose of study--------4
Significance of study-------4
Scope of study---------5
Review of related literature------5
2.1 Viewing conditions-------10
Observer and visual acuity------11
Those relating to the illuminator------14
2.2 Features of illuminator boxes------16
2.3 How to carry quality control test for illuminator---17
Direct assessment of surface perspex screen of illuminator--18
Assessment of the surface perspex screen of illuminator using black card. 19
2.4 Care of illuminators-------21
2.5Materials use in construction of illuminator box--21
2.6 Electrical components-------23
A rechargeable battery------24
Transformer -------- 35
Materials and methods------37
3.1 Preliminary steps-------37
3.2 Choice of design-------37
3.3Fabrication of the illuminator housing --- 38
Components and systems required: -----39
3.4 Machinery used in this contruction----39
3.5 Sources of materials------40
Other materials required and reasons------41
3.6 Fabrication procedure------42
Plate marking and cutting------42
Folding of plate--------44
Fabrication of the perspex screen.-----51
Materials used in the production of the screen---51
3.7 Voltage regulator circuit ------52
The stand and base-------54
4.1 Research findings/ quality control tests----55
4.1.1 Uniformity of the intensity of illuminator---55
Determination of the uniformity of the intensity of ilumination.-58
4.1.2Measuring the temperature of the surface over a period of time.------59
Quality control test for maximum discharging time--
Summary of results ------64
Recommendation from the study-----64
Area of further studies -------65
Limitations of study-------65
Radiographic viewing box is an illuminator use in viewing radiographs. Proper viewing conditions are very important when interpreting a radiograph. The viewing condition can enhance or degrade the subtle detail of the radiograph14. Before beginning the evaluation of radiograph, the viewing equipment and area should be considered. The area should be clean and free of distracting materials, magnifying aids, and film makers should be close at hand. Its function being to improves subject contrast of a radiographic image by enhancing the visualization of radiographs without which a radiographic image even with a good objectives contrast may not be well interpreted due to difference in perception of radiographic contrast by different observers9. X-ray illuminators have found increasing use in recent years as a tool for diagnosing medical problem.
Since the discovery of x-ray in 1895 by William roentgen in Germany, many x-ray equipment and accessories have evolved, all aiming at proper radiation protection and adequate diagnosis of illnesses. Among these inventions are the x-ray illuminators, which play an important role in film viewing, perception and interpretation of visible images.
Unfortunately, conventional x-ray illuminators lack recharge ability which is a factor that has been limiting its use in the event of power failure.
The aim of this project is to construct a battery powered dual face radiographic viewing box with rotatable neck and adjustable stand that alleviates the cumbersomeness of carrying a table to a long distance for viewing of radiographs.
This illuminator will improve grade performance and produce clear images for more simplified accurate diagnosis, although the x-ray viewing box is locally constructed but it will be international standard with many features like momentary start switch standard, internal ventilation with a fan high luster white color coat finish, quality steel material and will accommodate all kinds of x-rays without occupying too much space.
The average luminance is a better indicator of the viewing box luminance than the central luminance. This is confirmed by the greater standard deviation in the mean central luminance than the standard for the mean average luminance12. Screen-film technology still has widespread use in developing countries. As such it is imperative to have quality control measures on viewing boxes. Despite the widespread use of screen-film technology in developing countries, there is a gradual shift to digital X-ray systems, which could be the ultimate solution to non-optimal viewing box luminance having a detrimental effect on radiograph reporting. The use of digital systems paves the way for other viewing options like the use of video monitors, printing the image on paper and the use of picture archiving and communication systems (PACS), which have the advantage of eliminating the cost of film, chemicals and processor equipment. It should not escape one’s mind that digital imaging systems also have their relevant quality control requirements needed for optimal viewing of images.
For the development of world class Medical X-Ray Illuminators (View Box), we possess a robust infrastructure, which is spread over a vast area. It includes a design studio and a manufacturing unit that work together to ensure slimmest designs and high performance of our illuminators. All the components used in the fabrication process are sourced from quality conscious vendors. Our infrastructure also includes a quality testing facility, wherein the finished products are tested for their quality.
Local Construction Of Battery Powered Dual Face Radiographic Viewing Box With Rotatable Neck And Adjustable Stand
ABSTRACT The work was an attempt to evolve a quicker and easier means of reaching high places and point safely in tall buildings and other high structures. This has no doubt saved consideration cost of employing cranes. A very clear account of the introduction and literature background of the project has been discussed in details. The report also... Continue Reading
The focus of this project was the development of an ICU/Rehabilitation adjustable bed for the hospital and home environment. The project was initiated with the desire to develop an innovative, effective and inexpensive mechanism that would require an electrical supply. The design was performed systematically using various methods, procedures,... Continue Reading
ABSTRACT Cell phones need electric sources to charge their batteries in order to work, but there are people in developing and third world countries that find it hard to access electric sources.In an attempt to provide an alternative means of charging to those unfortunate areas in third world and developing countries, a solar powered cellphone... Continue Reading
ABSTRACT Cell phones need electric sources to charge their batteries in order to work, but there are people in developing and third world countries that find it hard to access electric sources.In an attempt to provide an alternative means of charging to those unfortunate areas in third world and developing countries, a solar powered cell phone... Continue Reading
ABSTRACT This project which is on the design and fabrication of a two face gas burner is a brief exploration on the various ways and means of designing, construction, operating the two face gas burner, putting cost into consideration for economic advantage. Herein are brief information and literature on gas... Continue Reading
Abstract In this project we designed a solar powered computer control system for electrical machines and spark ignition engines. It is made up of a hardware device that comprises of various electrical components which are powered by a 25W solar panel. The power cord of the machines are modified and connected to the hardware and the ... Continue Reading
ABSTRACT This project, 12v battery charge consist of few chapters, deal extensively with the design and construction of a batter charge which is capable of charging a 12v battery. The introductory chapter goes a long why in defining the battery charger and explanations of the mode of operations will be death with as we proceed. TABLE OF CONTENT... Continue Reading
ABSTRACT This project presents the design and construction of a battery charger. A battery charger is an electrical/electronic device used to put energy into a secondary cell or rechargeable battery by forcing an electric current through it. The system consists of a step down transformer, an AC to DC converter and a DC voltage regulator. The... Continue Reading
ABSTRACT The battery which has always serve as a sources of electrical current has in the past been invalid and inmost cases seen as monument after implicating as battery wound have to be brought over and over again the implicating has been that of lot of money is been spent for this purpose we give appraise for the advancement of technology which... Continue Reading
CONSTRUCTION OF A STANDARD BANKING OVEN (MINI-OVEN) WITH DUAL POWER SOURCE (ELECTRIC/GAS) ABSTRACT This project is designed to make the students utilize as many theoretical and practical concept as have been instructed in alstron to mechanical engineering workshop technology. Students are expected to learn and apply concepts, principles about... Continue Reading